Walnut Medical is funded by CAWACH (Center for War Against Covid) fund by Department of Science and Technology (Government of India) in collaboration with IIT Delhi.
Walnut Medical is proud to be India’s first Manufacturer of Oxygen Concentrators; two models are available 5 and 10 liter per minute flow rate.
Our Oxygen Concentrators boast a 93+-3% Oxygen Purity Level at Max Flow. We also use very high quality pumps that can maintain this level of oxygen purity over prolonged time periods while also withstanding voltage fluctuations and other nuances in our power supply.
Models Available: 1. MS-OC-05 (5 litre capacity) (Single and Dual flow) Download Brochure 2. MS-OC-10 (10 litre capacity) (Single and Dual flow) Download Brochure
Additional Documents: Pre-Compliance Testing Oxygen Purity Test Report
In the News Pharmabiz.com 11th June 2021 The New Indian Express 9th June 2021 DD News 11th June 2021 Latestly.com 9th June 2021 Yahoo! News 9th June 2021 Biospectrum 9th June 2021 Navbharat Times 9th June 2021 The Ken 31st May 2021
Walnut Medical Oxygen Concentrator Working Principle
The composition of air (78% Nitrogen, 21% Oxygen and 1% other gases like Carbon Dioxide, Argon, etc.) clearly shows that air is mainly comprised of two gases: Nitrogen and Oxygen [together 99%]. If Nitrogen is removed from air, the primary gas remaining would be Oxygen with purity of about 90-95%. Oxygen concentrators work on the principle of ‘Rapid Pressure Swing Adsorption’ which is where the Nitrogen is removed from the air using zeolite minerals which adsorb the Nitrogen, leaving the other gases to pass through and leaving oxygen as the primary gas.
Once the oxygen is collected the pressure then drops which allows Nitrogen to desorb and be expelled back into the air.
An oxygen concentrator has an air compressor, two cylinders filled with zeolite pellets, a pressure equalizing reservoir and valves and tubes. During the first half-cycle the first cylinder receives air from the compressor, which lasts about 3 seconds. During that time the pressure in the first cylinder rises from atmospheric to a few times normal atmospheric pressure (about 20 psi) and the zeolite becomes saturated with nitrogen. As the first cylinder reaches near pure oxygen (there are small amounts of argon, CO2, water vapour, radon and other minor atmospheric components) a valve opens and the oxygen enriched gas flows to the pressure equalizing reservoir, which connects to the patient’s oxygen hose. At the end of the first half of the cycle, the air from the compressor is directed to the 2nd cylinder. Pressure in the first cylinder drops as the enriched oxygen moves into the reservoir, allowing the nitrogen to be desorbed back into gas. Part way through the second half of the cycle there is another valve position change to vent the gas in the first cylinder back into the ambient atmosphere, keeping the concentration of oxygen in the pressure equalizing reservoir from falling below about 90%. The pressure in the hose delivering oxygen from the equalizing reservoir is kept steady by a pressure reducing valve.